For the 8200-series oscilloscopes+ what is horizontal resolution and how do I determine what it is?
The Horizontal resolution is depended on the Time per Division setting and the record length selected
The equation is Time per Division x 10 horizontal Divisions = Total Sample Interval x Record Length = Resolution per Sample Point. Or the easier method is use the scope menu Setup/Horizontal, Here you will find the resolution is listed.
Product: CSA8000B, CSA8000, TDS8000, TDS8000B, CSA8200, TDS8200
常见问题ID : 60616查看所有常见问题 »