Is the source isolated from the measurement circuitry?
Yes, there are no galvanic connections between the source and measurement circuitry. The sine wave source's circuitry is powered by its own transformer secondary and the control signals are optically coupled from the instrument's microprocessor circuitry. As a result, the source and measurement circuitry can be at different potentials and ground loops are eliminated. Thus, the 2015 prevents ground loop induced noise from coupling into the system.
常见问题 ID 71241