What are 'counts' in the accuracy specification?
Product(s) :Digital Multimeters: 175,175A,197,197A
Micro-ohm Meter: 580
CV/IV Meters: 595,590
Low-Current/High-Resistance Products: 6514, 6517, 6517A, Model 65
Question : What are â€˜countsâ€™ in the accuracy specification?
Answer: Unless stated otherwise, 1 count is equal to 1 unit of resolution of the selected range
The two terms in the specification are the zero and gain terms. Where % of reading is the gain term and the counts is the zero or offset term.
Here is an example of how to calculate accuracy when measuring a 0.5V signal on the 1V range when that range has a 10uV resolution.
The specification is stated as: 3% + 5 counts
This translates to 3% * (signal input) + 5 * (resolution of that range)
0.03*0.5 + 5*10uV = .015V + 50uV = 0.015050 = 15.05 mV
For the applied 0.5 volts, the reported measurement will be within specification as long as it is in the range of .48495 and .51505
常见问题 ID 69831查看所有常见问题 »